Radar level transmitter wholesale from China? The key components are made of high-quality materials, which have strong corrosion resistance and can adapt to highly corrosive environments. Low power consumption, can use solar power to supply power, no need to build water level wells, adapt to various geographical environments, no impact on water flow, and more convenient installation and maintenance. The parameter setting is convenient, and the false echo from the liquid surface to the antenna can be automatically identified by the software carried by itself to eliminate the interference. Read more information on radar level measurement.
For the ultrasonic instrument that continuously measures the liquid level, when the temperature of the liquid to be measured changes greatly, the change of the propagation speed of the sound wave should be compensated. The connecting cable between the detector and the converter should take anti-electromagnetic interference measures. The structure of the ultrasonic level sensor should be determined according to the process requirements and on-site working conditions.
In the process of cement production, material level detection and material level alarm control play an important role from mining and crushing to storage and packing of finished cement. But because of the complex working conditions of cement plant, radar level meter has no small difficulty in measuring. Today, I would like to talk with you about specific difficulties and solutions. First cement plant material for measurement in the dust under the condition of particularly large, especially some pneumatic feed bin, involves the kinds of materials, material particle size is differ, dielectric constant is different also, such as broken limestone, raw meal, clinker, fly ash, coal, cement, the level of the bunker, it is very difficult to accurately measure Especially fly ash bin with low dielectric constant.
Rod antenna: generally used in strong corrosive environments, with weak anti-interference ability and small range; Flare antenna: stronger anti-interference ability, suitable for more complex environments. The larger the bell mouth, the more concentrated the energy, and the larger the measuring range; Parabolic antenna: the focusing effect is stronger than that of the bell mouth, the anti-interference ability is the strongest, and the range is the largest.
Mechanical interference, the so-called mechanical interference, means that some components inside the radar level sensor will also vibrate due to the large-scale vibration or impact of the external machine, and even undergo displacement and deformation, which may also cause the pointer of the instrument head to loosen and cause measurement. error. In this case, we usually use spacers, shock springs, etc. to cushion the shock.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read additional details at https://www.kaidi86.com/. The Magnetic Level Gauge all use vacuum tube technology, with a lifespan of 3-5 years, and protection grade is up to IP68, not easy to fade.
There is AC interference and the voltage is high. For example, for the radar level meter used in the production line, the power supply requirement is 24VDC (typical value), but in the on-site measurement, it is found that the power supply is displayed as 27.2V, which is significantly higher than 24VDC, resulting in a large measurement result and even a radar level meter. crash phenomenon. The installation position of the radar level meter is incorrect, which leads to deviations in the measurement. For example, the accumulation of aggregates in the transfer bin is a “mountain”-shaped cone, but only one radar level meter is installed near the discharge port of the return belt. , the installation position is too close to the discharge opening of the return belt, and at the same time, it is too far from the discharge opening of the feeding belt on both sides. Just below the radar level meter is the drop point of the return belt. If the distance is too close, the aggregate in the falling process will interfere with the radar level meter and form false reflections.