Best level indicator manufacturer? For radar level gauges, there are many reasons for interference and many sources of interference. We analyze from four aspects: internal, external, AC and DC. Celestial and celestial interference, first of all, what is celestial interference? Celestial objects refer to the sun or other stars, therefore, celestial interference refers to the interference of their electromagnetic waves on the radar level gauge. We are very unfamiliar with Tiandian. The so-called Tiandian is usually understood as the interference of the signal of the magnetrol radar level gauge caused by the ionization of the atmosphere, lightning, or the electromagnetic waves generated by natural phenomena such as volcanoes and earthquakes. Discover extra information on https://www.kaidi86.com/product.html. OEM Service-can be customized according to the application of customer,such as level gauge,level switch and level indicator.
The installation of the liquid level gauge should be away from the inlet, outlet, eddy current or protrusions on the inner wall to avoid the interference of electromagnetic waves and affect the measurement. When installing the guided wave radar, make sure that the diameter of the short pipe at the flange connection is greater than the height of the short pipe, otherwise the cable will contact the short pipe wall due to shaking, and the liquid level cannot be measured accurately. How to choose a model, in the final analysis, you still need to understand your own working conditions and the characteristics of different instruments, and judge whether the two match, so that you can get twice the result with half the effort.Want to know about ultrasonic flow meter advantages and disadvantages, click here.
Power supply and output signal, the power supply has 220 V A C, 24 VDC, and the two-wire or four-wire system can be selected according to the needs. Output signal 4 ~ 20 mA DC or digital signal, select the appropriate model according to the needs. The above are all factors that need to be considered when selecting a radar level transmitter. The essence of the details is a serious attitude and a scientific spirit. I hope that every user can choose a suitable radar level meter.
There is AC interference and the voltage is high. For example, for the radar level meter used in the production line, the power supply requirement is 24VDC (typical value), but in the on-site measurement, it is found that the power supply is displayed as 27.2V, which is significantly higher than 24VDC, resulting in a large measurement result and even a radar level meter. crash phenomenon. The installation position of the radar level meter is incorrect, which leads to deviations in the measurement. For example, the accumulation of aggregates in the transfer bin is a “mountain”-shaped cone, but only one radar level meter is installed near the discharge port of the return belt. , the installation position is too close to the discharge opening of the return belt, and at the same time, it is too far from the discharge opening of the feeding belt on both sides. Just below the radar level meter is the drop point of the return belt. If the distance is too close, the aggregate in the falling process will interfere with the radar level meter and form false reflections.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read more information on kaidi86.com.